22 May / June 2021 Retention Time Variability
Changes in the retention times of target analytes can lead to incorrect identifi cation especially when using non-selective detectors such as a Flame Ionisation Detector (FID).
a) Leaks in any part of the system can affect the fl ow of the carrier gas which will cause variability in the retention time of the target analytes. Leak check the inlet and any column connections.
b) Replace parts such as the septa and O-rings if they show signs of wear. 2. Analyte adsorption
a) Increased activity in the inlet and column may result in adsorption of the target analyte. This may affect its transfer to the column and retention on the column. This can be overcome by routine inlet and column maintenance.
b) Use of properly deactivated liners and columns will reduce the potential for active sites.
3. Resolution and integration issues.
a) Adding too much sample can overload the column. This can cause variability in the retention time as well as poor resolution and peak fronting. This can be avoided by diluting the sample, reducing the injection volume and/or increasing the split ratio.
4. Incorrect column/oven temperature program
a) If the column temperature and the oven temperature program are not appropriate for the analysis, there may be variability between runs. Verify that the column temperature and oven temperature program are optimised and set correctly in the acquisition method.
5. Incorrect or variable carrier gas linear velocity.
a) The retention times of the target analytes may be affected by variations in the linear velocity of the carrier gas. Verify the carrier gas linear velocity is optimised and set correctly in the acquisition method.
b) If the instrument is unable to maintain the correct linear velocity, there may be a problem with the fl ow controller or a leak in the system. Check the system is functioning correctly and repair or replace parts if necessary.
6. Poor control of oven temperature programming.
a) If the oven temperature program in the method is outside the capability of the instrument, it will be unable to accurately maintain the desired temperature. This can cause variability between runs and affect the analyte retention times. This can be resolved by confi rming GC oven program falls within instrument specifi cations.
7. Incorrect oven equilibration time.
a) The oven may not have enough time to stabilise before the run starts. This can be more of an issue if the oven temperature at the start of the run is close to room temperature. This can be overcome by extending the GC oven equilibration time in the acquisition method.
8. Manual injections.
a) When carrying out manual injections, inconsistencies between the injection and pressing the start button can lead to retention time variability. Where possible, standardise the manual injection procedure. Alternatively, using an autosampler will improve reproducibility between injections.
These systems, like all chromatographic techniques, have a list of quite common problems seen by analysts. To aid your troubleshooting, Shimadzu have developed a free troubleshooting guide which discusses these common problems and a list of helpful actions that can be taken. It offers guidance on how to approach certain problems in a methodical, scientifi c manner. Patience is required and don’t be disheartened if your fi rst course of action doesn’t resolve the issue. Remember that the most obvious answers are usually the correct ones, and there is nothing wrong with checking the basics fi rst.
Finally, if you do have a problem please remember to call your instrument provider. Ultimately, they are here to help, and all want to see instruments up and running as much as you!
The free trouble shooting guide is available from: www.sh
Affordable Single Plate Evaporator Launched Porvair Sciences has launched its new generation Ultravap®
Levante nitrogen blowdown sample evaporator. The second generation system offers the many benefits associated with Porvair’s flagship evaporator, the Ultravap® Mistral, in
an affordable, easy-to-use package. This state-of-the-art single microplate evaporator is designed to meet the current needs of most chromatography laboratories whilst providing the future option of fully automating sample dry-down in applications such as LC/MS analysis, dry recovery or reconstitution.
The compact, standalone Ultravap® Levante is fully programmable and easy to control from a robot liquid handler. Its
advanced head technology enables delivery of evaporation temperatures up to 80°C for faster solvent evaporation rates. If required the system has the option to be fitted with ducted fan fume extraction.
Operated via an intuitive graphical colour LED touchscreen, software features includes up to 5 stored alphanumerically named programmes and multi-step evaporation programmes. This functionality combined with real-time run displays showing actual gas temperature, gas flow rate and stage height means the Ultravap®
acrylic splash guard safeguards the operator and also ensures efficient removal of the solvent vapour from the evaporation table. The Ultravap®
Levante puts you in full control of the evaporation process. A removable
The new generation Ultravap®
Levante is supplied with an evaporator head of your choice, which may be interchanged quickly and easily with a head of a different pattern as your workflow demands. The precision engineered mechanism uses a standard ANSI/SLAS plate nest to accept most microplate formats and tube racks. It can accommodate tubes up to 50mm in height in a variety of configurations to allow the use of 2 Dram vials, 1.5ml HPLC vials, bar-coded tubes in racks and many other common formats. Evaporator heads are available for efficient dry down of 24-, 48-, 96- and 384-well microplates at temperatures up to 80°C
Levante can be operated from both a gas cylinder and an in-house supply of nitrogen or clean dry air. The blow-down evaporation technique is well proven for efficiently removing chromatography solvents such as dichloromethane, acetonitrile, methanol and hexane.
More information online: ilmt.co/PL/3dAw
nitrogen blowdown sample evaporator.
| Page 2
| Page 3
| Page 4
| Page 5
| Page 6
| Page 7
| Page 8
| Page 9
| Page 10
| Page 11
| Page 12
| Page 13
| Page 14
| Page 15
| Page 16
| Page 17
| Page 18
| Page 19
| Page 20
| Page 21
| Page 22
| Page 23
| Page 24
| Page 25
| Page 26
| Page 27
| Page 28
| Page 29
| Page 30
| Page 31
| Page 32
| Page 33
| Page 34
| Page 35
| Page 36
| Page 37
| Page 38
| Page 39
| Page 40
| Page 41
| Page 42
| Page 43
| Page 44
| Page 45
| Page 46
| Page 47
| Page 48
| Page 49
| Page 50
| Page 51
| Page 52
| Page 53
| Page 54
| Page 55
| Page 56
| Page 57
| Page 58
| Page 59
| Page 60