Meeting the needs
days in order to reduce costs. The SmartSite®
fulfil these requirements (ensuring the sterile state and operator safety). As well as minimising the risk of accidental contact by the operator with the drug, these CareFusion devices also make it possible to achieve quickly and simply a closed system, which is indispensible for guaranteeing the administration of a microbiologically inert preparation to the patient. SmartSite®
is a vented vial access
is a closed male valve for connection between the SmartSite®
device, equipped with a 0.22µ filter; Texium®
luer-lock syringe. Two significant risk factors in the preparation of these solutions are aerosol contamination due to needle sampling and needle stick injury to the operator. These two devices allow needle-free sampling and eliminate contamination. Both devices are equipped with nondrip technology and with an automatic locking system upon disconnection, and easy and safe connections between the parts. The above-mentioned devices are shown in Figure 1. They are equipped with an internal plunger lubricated with medical-grade silicone oils that permit the parts inside the device to slide. They are made of polyurethane, lipid-resistant polycarbonate and silicone polymers (internal plunger). Their combined use constitutes a closed system that prevents contamination and leakage of any solution being handled.
The compatibility of these products with various solvents and drugs has already been tested during the approval phase; in our study, the investigation of the compatibility of these devices was also extended to cover the evaluation of the possible interaction that might occur with any antiblastic drugs and the verification of the actual in-use stability of the main products on the market in order to facilitate reuse and the management of production residues, thus generating considerable cost savings.
The work was carried out within the context of a collaboration between the University of Siena and the Siena University Hospital Pharmacy and stems from the latter’s need to provide themselves with their own validated methods for managing AC drugs, with particular regard to the possibility of reusing the daily residues resulting from the preparation of the ACs themselves. Where necessary, the pharmaceutical forms were reconstituted at the Siena
University Hospital galenic laboratories. Trastuzumab (Herceptin® bortezomib (Velcade®
were selected as being representative of the monoclonal antibodies; the logic behind this choice was that both occur as lyophiles to be reconstituted and thus present greater stability problems compared to the ready-to-use solutions for which only samplings need to be performed. Docetaxel and paclitaxel, of which the following speciality medicinal products were tested, were selected from among the drugs formulated in more aggressive solvents (ethyl alcohol): Docetaxel 20mg/ml (Actavis), Taxotere® 10mg/ml (Sanofi Aventis) and Anzatax® 6mg/ml (Pharmaplan).
All of the specialty medicinal products were sampled on the scheduled days and then diluted with super-pure water prior to the LC-UV chromatographic analysis. The stability studies were carried out following the development and validation of appropriate chromatographic methods.
Stability tests in vial
The solutions of the various preparations were kept in the refrigerator in the dark and at ambient temperature (light and dark) with the SmartSite® valve inserted. The content of the solutions of the various drugs remains constant during the seven days of testing. The LC-UV chromatographic profiles do not indicate the presence of chromatographic peaks ascribable to possible degradation products.
Stability tests in syringe For this purpose, the solutions of the various preparations were kept in the refrigerator in the dark and at ambient temperature (light and dark) in a polypropylene syringe with the Texium® valve inserted. The tests performed and the substantial similarity of the previously obtained results can lead us to conclude that the drugs are stable over the seven test days and that neither the materials used nor the sampling devices nor the environmental conditions cause significant variations in the content of the latter.
The sampling devices were manufactured in such a way as to maintain the conditions of sterility of the solutions contained in them and the manufacturer provides data regarding functionality tests on the silicone plunger located inside the device itself. These tests, performed with various solutions
containing different kinds of drugs (for example fluorouracil, glycerine and insulin), have verified that the device does not suffer damage and maintains its functionality.
The specific migration tests carried out showed that there are no significant migration phenomena under the conditions used with the LC-MS detection (water, acidic or ethanolic solutions). Differences were not observed in any case between the chromatographic profiles of the solvent used before and after prolonged contact with the devices. With regard to the medical-grade silicone oils, used as lubricants in the part of the plunger using alcoholic solutions, their extractable quantity with ethyl alcohol was determined by comparison with special calibration curves set up by analysing solutions at scalar concentration. The quantity of silicone oils extracted from the devices during the usual practice of the preparation of the drugs in alcoholic solution were below the detection limits of the GC-MS method (12µg/device).
The results showed that the analysed solutions are stable during the seven days of testing, since their content never goes below 96%.
The tests carried out and the
substantial similarity with the contact with the glass lead us to conclude that the material the sampling devices are made of in no way affects the stability of the selected drugs during the sampling phases and/or direct contact with the same. We also checked the possible migration by the sampling devices using various solutions and various analytical techniques (LC-MS and GC-MS). Peaks were not recorded in any case using the LC-MS technique in the tested solutions. From the results obtained with GC-MS, it was possible to establish a quantity of silicone oils extractable from the devices during the usual practice of the preparation of the drugs in alcoholic solution.
The reported data represent a real help to hospital pharmacists deciding to work according to the principles of in-use stability. These results make it possible to reuse the production residues, which would otherwise need to be discarded. This can lead to significant savings, which, during times of economic downturn, can make the system more sustainable. l
| Page 2
| Page 3
| Page 4
| Page 5
| Page 6
| Page 7
| Page 8
| Page 9
| Page 10
| Page 11
| Page 12
| Page 13
| Page 14
| Page 15
| Page 16
| Page 17
| Page 18
| Page 19
| Page 20
| Page 21
| Page 22
| Page 23
| Page 24
| Page 25
| Page 26
| Page 27
| Page 28